ICEweb has nearly 100 Control, Instrumentation, Fire & Gas, Safety Instrumented Systems core pages and a total of more than 300 pages - It Really is Cool Engineering - By Engineers for Engineers it must be just about the World's first choice for Technical Information.
Whilst every effort is made to ensure technical accuracy of the information supplied on iceweb.com.au, Keyfleet Pty Ltd and its employees accept no liability for any loss or damage caused by error or omission from the data supplied. Users should make and rely on their own independent inquiries. By accessing the site users accept this condition. Should you note any error/omission or an article offends please do not ignore it, contact the webmaster and we will review, rectify and remove as necessary.
Get seen by the people who use your
can be yours
Equal Column Liquid Manometer
This is one of the oldest methods of measuring differential pressure. In its most basic configuration it consists of two identical vertical tubes partly filled with a liquid. These tubes are connected together at the bottom. Pressure is measured by determining the change in height between the two tubes and is scaled according to the density of the liquid. Thus different fluids will give a change in the pressure range.
Typical fluids used are water and mercury.
The method is very accurate however is limited to lower ranges of pressure. It is rarely used today, however it used to be used quite extensively as a calibration standard for low pressure instruments and control valves. This role has now been taken over by highly accurate calibration tools. The following diagram depicts an equal column liquid manometer. It shows a glass tube bent into a "U" shape. This tube is partly filled with the fluid and a pressure scale is located between both tubes.
The Principle of an equal
column "U" tube manometer
Page 2, Close